Output driver nonrepetitive peak on-state current (tw = 10 ms, duty cycle = 10%, see Figure 7) . . . . . . 1.2 A
3. Operating Instructions
3.1 . Operation steps
3.1.1 First the 5 m extension cable, splitter, thermocouple temperature probe connected and inserted to LAN2thermoouple interface, connect network Cable .please ensure that connect normal.
3.1.2 Turn on the power .LAN port of LINK light blink, flashing that said ACK network connection to normal. If not, please check your network is normal or not. Red LED the unusual light flashing, said equipment find probe, and reading temperature. If not, please check your probes whether good connection, you can connect well probe again after restart your device.
3.1.3 Open software, if your IP section is not our default 192.168.1. XXX section, please manually add own IP section, (see picture 3-2-1 and 3-2-2)
3.14 Add IP successful, and then open the software can find equipment if still can't find your IP that prove setting wrong or subnet mask setting wrong. When you find device, then you can modify the device's IP.
3.15 At this time, the normal work equipment, if temperature display “error” , that says the probe didn't insert or contact undesirable, please adjust the probe .
3.2 Frequently Questions and resolution
1 Q: why Equipment connected not success?
A: Checking Network Cable connected and checking your LAN IP segment. If continue does not work, Please press the reset button, IP restore the factory settings.
2 Q: why the device cannot find the probe?
A: whether it is operational problem? Please ensure that all equipment with a good, then, connect the power, the device will immediately search the probe. So, please check whether the probe is connected short circuit or poor contact. Because this is a line of communication, if there is a probe connect poor, all the other probes would happen error.
3 Q: Why read the temperature will be ERROR?
A: This usually is because the probe cable is too long, resulting in the probe power supply shortage.
4 Q: Why does the device will be dead running some days later?
A: 1. this equipment should not be on the high temperature and high pressure areas. 2. Part of Probe connection has short circuit problem.
5 Q: Why the detected temperature will become small when the environment temperature become high?
A: Thermocouple probe “+” and “-” the wrong position, the letter “K” must face upward.