(6) Inner of its have Temperature warning settings . such as :upper and lower limit of temperature warning setting .
5.Installing instruction: (Note: Our software only support windows system .)
you need to install below drivers while you install the software at first time.
a) install correspond system's driver , 32-bit systems with x32, 64-bit systems with x64 .
(2) 1 Wire Driver
a)install correspond system's driver , 32-bit systems with x32, 64-bit systems with x64 .
(3) Install the application: Double-click setup.exe of the root directory to install the software
a)plug TEMPer18B20 into the computer and running TEMPer1 w, thenstart to measure the temperature. See the picture below:
Picture1 Read the real time temperature
This option allows you to settingtemperature of any channel , for example: modify the calibration value, upper ,lower and address of the alias, click "OK" button. If you connect a lot of measuring points, the software did not timely identify them, you can click "refresh device",measurement points have not been identified will be displayed in the software.
b) Loading the curve graph: You can load the curve graph on the “Curve” option part. Note, the device must be connecting to the computer to work, so the curve graph is the one about the real time temperature. And the function button can move, magnify, print, copy and save the curve graph. When you connect hundreds of test points, because the software interface is limited, can not be all of the test point curves show at the software . At this time, you can click on: "edit", select you want to show the test points
Picture 2 Curve graph
c)Setting :In the “setting” option you can transfer the deferent measuring units, the reading interval , the language and save file settings .and set a upper and lower limit,
Picture 3 Setting
d)Alarm : in temperature alarm settings option , you can choose different alarm mode ;in “Email configuration” , you can send the temperature records to your mailbox. Before to do this, you must set the server of the mailbox, the mail address of sender and receiver, user’s name and so on. And you also can set the sending conditions and interval in the option.
Picture 4 Alarm setting
e) About Options: Display your current software version.
Picture 5 About
if you don't want to use our software ,you also can use maxim One Wire Viewer software .